|Reduction of implant RF heating through modification of transmit coil electric field|
Y Eryaman, B Akin, E Atalar
Magnetic resonance in medicine 65 (5), 1305-1313, 2011
|Parallel transmit pulse design for patients with deep brain stimulation implants|
Y Eryaman, B Guerin, C Akgun, JL Herraiz, A Martin, A Torrado‐Carvajal, ...
Magnetic resonance in medicine 73 (5), 1896-1903, 2015
|Toward imaging the body at 10.5 tesla|
MA Ertürk, X Wu, Y Eryaman, PF Van de Moortele, EJ Auerbach, ...
Magnetic resonance in medicine 77 (1), 434-443, 2017
|Reduction of the radiofrequency heating of metallic devices using a dual‐drive birdcage coil|
Y Eryaman, EA Turk, C Oto, O Algin, E Atalar
Magnetic resonance in medicine 69 (3), 845-852, 2013
|Fast electromagnetic analysis of MRI transmit RF coils based on accelerated integral equation methods|
JF Villena, AG Polimeridis, Y Eryaman, E Adalsteinsson, LL Wald, ...
IEEE Transactions on Biomedical Engineering 63 (11), 2250-2261, 2016
|Miniaturized fiber‐optic transmission system for MRI signals|
OG Memis, Y Eryaman, O Aytur, E Atalar
Magnetic Resonance in Medicine: An Official Journal of the International …, 2008
|Evaluation of internal MRI coils using ultimate intrinsic SNR|
H Çelik, Y Eryaman, A Altıntaş, IA Abdel‐Hafez, E Atalar
Magnetic Resonance in Medicine: An Official Journal of the International …, 2004
|Transcranial focused ultrasound for BOLD fMRI signal modulation in humans|
L Ai, JK Mueller, A Grant, Y Eryaman, W Legon
2016 38th Annual International Conference of the IEEE Engineering in …, 2016
|Magnetic resonance method and apparatus for reducing RF heating in the patient|
Y Eryaman, E Atalar
US Patent 8,319,496, 2012
|SAR reduction in 7T C‐spine imaging using a “dark modes” transmit array strategy|
Y Eryaman, B Guerin, B Keil, A Mareyam, JL Herraiz, RK Kosior, A Martin, ...
Magnetic resonance in medicine 73 (4), 1533-1539, 2015
|Multi‐atlas and label fusion approach for patient‐specific MRI based skull estimation|
A Torrado‐Carvajal, JL Herraiz, JA Hernandez‐Tamames, ...
Magnetic resonance in medicine 75 (4), 1797-1807, 2016
|Design of internal MRI coils using ultimate intrinsic SNR|
Y Eryaman, Y Öner, E Atalar
Magnetic Resonance Materials in Physics, Biology and Medicine 22 (4), 221-228, 2009
|A simple geometric analysis method for measuring and mitigating RF induced currents on deep brain stimulation leads by multichannel transmission/reception|
Y Eryaman, N Kobayashi, S Moen, J Aman, A Grant, JT Vaughan, ...
NeuroImage 184, 658-668, 2019
|First in‐vivo human imaging at 10.5 T: Imaging the body at 447 MHz|
X He, MA Ertürk, A Grant, X Wu, RL Lagore, L DelaBarre, Y Eryaman, ...
Magnetic resonance in medicine 84 (1), 289-303, 2020
|Radiofrequency heating studies on anesthetized swine using fractionated dipole antennas at 10.5 T|
Y Eryaman, RL Lagore, MA Ertürk, L Utecht, P Zhang, A Torrado‐Carvajal, ...
Magnetic resonance in medicine 79 (1), 479-488, 2018
|Combined loop+ dipole arrays for 7 T brain imaging|
Y Eryaman, B Guerin, R Kosior, E Adalsteinsson, LL Wald
Proc 21st Annual Meeting ISMRM, Salt Lake City, 2013
|A method for phantom conductivity and permittivity measurements|
|In vivo human head MRI at 10.5 T: A radiofrequency safety study and preliminary imaging results|
A Sadeghi‐Tarakameh, L DelaBarre, RL Lagore, A Torrado‐Carvajal, ...
Magnetic resonance in medicine 84 (1), 484-496, 2020
|Optimizing the topography of transmit coils for SAR management|
A Sadeghi-Tarakameh, A Torrado-Carvajal, C Ariyurek, E Atalar, ...
Proceedings of the 26th Joint Annual Meeting of ISMRMESMRMB, Paris, France, 2018
|MRI Compatible pacemaker leads|
O Ferhanoglu, Y Eryaman, E Atalar
Proc Intl Soc Mag Reson Med 13, 963, 2005